A mains electricity powered device that concentrates oxygen from ambient
air and delivers the concentrated oxygen in a controlled manner to a
patient requiring oxygen therapy via a single outlet up to 10 L/min.
Oxygen concentrator, 10 LPM – single flow
Oxygen concentrators are not suitable to be used as an oxygen source to
other medical equipment providing respiratory support, such as
ventilators, CPAP devices, High Flow Nasal Oxygen, etc.
Provides a continuous, variable flow of concentrated oxygen (>82%)
derived from room air, to a maximum rate 10 L/min.
Contains oxygen monitor to verify concentration.
Equipped with one oxygen outlet, provided with controllable flow meter.
Audible and visual alarms for low oxygen concentration (<82%) and powersupply failure.
Audible and visual alarms for high temperature, no flow rate and
Power efficiency < 70 W/L/min.
Digital meter that displays cumulative hours of device operation.
Oxygen outlet(s) with 6 mm (¼-inch) barbed fitting and DISS connector.
Flowmeter minimum flow rate of 2 L/min. Lower flow rates may be achieved
by a combination of the concentrator and a separately supplied flow
splitter (details under RELATED PRODUCTS).
Flowmeter continuously adjustable, with markings of at most 0.5 L/min
Flowmeter control allows shutting off flow (i.e. 0 L/min) by user.
Noise level <60 dB(A).Casing and environment
Hard case, cleanable with standard hospital cleaning materials.
Whole unit movable with wheels on at least two feet
Unit weight <27 kg
The unit includes internally and externally mounted filters for cleaning
the air intake.
All user-removable filters are cleanable. Cleaning instructions for
filters are included in the instructions for use.
Electrical power requirements are locally compatible.
230 Volt, 50 Hz, 3.0 amps (120 Volt units available)
Mains power cable length > 2.5 m.
Capacity for safe operation from at least ± 10% of rated voltage.
Electrical protection by resettable circuit breakers, fitted in both
neutral and live lines.
ITEMS SUPPLIED WITH
2 x adult cannula with 2m kink-resistant oxygen tubing with standard
4 x paediatric cannula with 2m kink-resistant oxygen tubing with
1 foam filter
1 felt filter
DISS to 6mm barbed adaptor for each outlet (if relevant)
1 x set user and maintenance manuals, hard and soft copies, in local
language, or in agreed other language if local is not available.
1 x certificate of calibration and inspection.
1 x list of equipment and procedures required for local cleaning,
disinfection, troubleshooting, calibration and routine maintenance.
1 x list of all spares and accessories, with part numbers and contact
details for parts supply.
1 x document with contact details of manufacturer, supplier and local
ITEMS REQUIRED, BUT NOT SUPPLIED
S0002640, Oxygen Analyzer, electrochem, handheld
S0002639, Oxygen Analyzer, ultrasonic, handheld
- Temperature: 10 °C to 40 °C,
- Relative humidity: 15% to 90%
- Elevation above sea level: 0 to 2000m
- Temperature: 0 °C to 40 °C,
- Relative humidity: 15% to 95%
- Elevation above sea level: 0 to 2000m
ESTIMATED LEAD TIME
4 – 6 weeks
The User Manual supplied with the device details all installation which
can be undertaken by users.
USER TRAINING REQUIREMENTS
The User Manual supplied with the device details all operation
procedures for user.
TECHNICAL TRAINING REQUIREMENTS
The User Manual supplied with the device details all maintenance and
troubleshooting procedures which can be undertaken by users.
WEIGHT AND VOLUME
Weight: 29.9 kg (packaged)
Volume: 0.108 m3 (packaged)
S0845014, Pulse oximeter,portable,w/access
S0002073, Uninterruptible power supply, 1kVA
S0845039, Flow splitter,for oxygen concentrator
S0845017, Pulse oximeter, handheld, incl.resp. rate
S0002624, Pulse oximeter, tabletop
S0002632, Voltage stabilizer, 1 kVA
S0002635, Spares/tool kit for oxygen concentrator
S0845049, Nasal Catheter, 8 Fr
S0845043, Tubing, oxygen supply, 7.6 mtr
S0845008, Oxygen concentrator, 8LPM, double flow
S0845041, Oxygen concentrator, 5 LPM
S0845033, Oxygen concentrator, 10 LPM, dual flow
S0845044, Oxygen concentrator SET
PACKAGING / LABELING:
Unit presentation: 1 (one)
Labelling on primary packaging must include:
- Name and/or trademark of the manufacturer
- Manufacturer's product reference
- Type of product and main characteristics
- If the packaging is not transparent, it must bear a diagram showing
the essential parts of the product
- Information for particular storage conditions (temperature, pressure,
light, humidity, etc.), as appropriate (or equivalent harmonised symbol)
- Information for handling, if applicable (or equivalent harmonised
Over packaging: Packaging unit
Labelling on the packaging unit: Labelling to be the same as primary
Extra information required: Number of units per box
Only logged in customers who have purchased this product may leave a review.
- Store Name: UNICEF Supply Division
- Vendor: Claudia Lago
- Address: Oceanvej 10-12
- No ratings found yet!